4.2.2 Do not open the lid of Autoclave, when cycle is running.
4.2.3 Take care during handling of the articles, as they will be hot, after sterilization.
4.2.4 After completion of cycle, open the lid of Autoclave, when the pressure gauge shows “0” reading and temperature gauge hows below 90 OC.
4.2.5 Ensure heating coils are completely submerged in Distilled water.
Promote this Post on your site,replace this code anywere you like.
Author: ichapps 184 days ago on Category: Health Care autoclave microbiology anticlockwise All